GEMINI™ 5540A RF Power Meter High Power, High-Accuracy RF Measurement in a Single Instrument
0.2 to 200 MHz, 3 to 5,000 W
Designed from the same dedication and expertise that built the TEGAM calorimetric instruments used in primary and production labs around the world, TEGAM’s new GEMINI 5540A RF Power Meter1 brings modern functionality and easy-to-use design to the world of in-line RF power measurement. Ideal for use in semiconductor applications where measurement uncertainty and repeatability are imperative, the GEMINI 5540A RF Power Meter is a measurement tool you can trust.
The 5540A is suitable for the most demanding applications with a 2-sigma accuracy of ±0.5% + 0.5 W across its power range. It combines the sensor and meter into one convenient instrument, allowing technicians to read forward and reverse power, VSWR, and frequency simultaneously on the large, easy-to-read OLED display.
It is powered by a standard USB 2.0 port, external battery, or AC power outlet using off-the-shelf USB adapters. And since the meter is built-in, there is less equipment – and less cables – to carry to the field or bench.
Real-time readings and data collection increase user productivity and process efficiency. Use TEGAM’s GEMINI Tools RF Power Meter Application, or write custom applications with the 5540A’s command set.
1 Patent No. 10,833,457
The GEMINI 5540A is available with a variety of connector types. When ordering, please identify the desired connectors as illustrated below:
Example: 5540A-Nf-Nm (GEMINI 5540A with Type N female input, and Type N male output)
- Accuracy: ±0.5% + 0.5 W at 2σ2
- 0.2-200MHz, at 3 to 5,000 W
- Read forward and reverse power, VSWR, and frequency simultaneously on the integrated display
- Insertion loss < 0.05 dB
- USB-2 power and data interface
- The GEMINI 5540A can be calibrated to either the input or output port, at one or more frequencies. Please contact TEGAMSales@AEI.com to discuss your calibration requirements.
2 Accuracy specification valid at calibrated frequencies, from 18 to 28° C. Specifications valid only when used with the connectors in-place at time of calibration. See GEMINI 5540A Datasheet for detailed specification information.